View and Download HP A manual change online. Impedance/Gain-Phase. A Multimeter pdf manual download. Hp A Pdf User Manuals. View online or download Hp A Manual Change. Current Directory: /HP Agilent/HP A Impedance Gain Phase Analyzer. Click on a File to download or a Directory to open. File Type. File Size. ===> Back.
|Published (Last):||26 May 2010|
|PDF File Size:||11.90 Mb|
|ePub File Size:||20.97 Mb|
|Price:||Free* [*Free Regsitration Required]|
Message 18 of You could use the HP only, set the bias voltage to 2. Detailed impedance and transmission characteristics, including secondary parameter derivations, can be simply and quickly evaluated or tested. Realistic device characteristics can be analyzed under actual operating conditions by varying the test frequency, test signal level, and dc bias.
Losses in inductor of a boost converter 9. The free PDF on Agilent’s web site has unreadable schematics. ModelSim – How to force a struct type written in SystemVerilog? Measuring air gap of a magnetic core for home-wound inductors and flyback transformer 7.
Without getting into the particular details of the experiment, an AC field is applied to the electrodes using a function generator, to form “wires” between the electrodes bridging the gap.
Heat sinks, Part 2: Can you majual a DC 2.
HP A MANUAL CHANGE Pdf Download.
How do you get an MCU design to 1494a quickly? Manuql Technology file 1. Mwnual problem is that we have to manipulate the circuit element with an external AC field via a function generator and simultaneously measure the circuit properties of the active nanostructure as a function of frequency using the impedance analyzer.
I am not sure that there is a way where you can separate the interference of the sine function from the Imp analyzer and the constant 2MHz from your FG. I am not going to claim to be an expert on impedance measurements, but I will try to help.
Using HP A’s Equivalent Circuit Analysis Function, you can easily and quickly obtain those equivalent circuit constants that, until now, required a number of time-consuming, complicated calculations.
PNP transistor not working 2. I chose this frequency range because the device behaves like a resistor in this frequency range. The time now is Message 16 of Message 20 of The analysis function not only provides you with impedance and transmission characteristics, but also allows ,anual to determine secondary parameters. Most Active Hardware Boards: Quadtech also has some good tech notes on impedance measurements. Hierarchical block is unconnected 3.
Manuals – KO4BB
HP A service manual 0. You can use the HP A’s powerful analysis functions to analyze these calculated parameters. Turn on power triac – proposed circuit analysis 0. High accuracy and wide range- Impedance measurement: Manuall modulator in Transmitter what is the A? The external AC field used to form these wires are 2.
As a result, we cannot use the HP analyzer as a function generator to manipulate the device and also do a frequency sweep to measure the properties at the same time. What is the function of TR1 in this circuit 3.
Distorted Sine output from Transformer 8. Message 17 of Added after 11 minutes: I had a quick question on making impedance manul of a device using the HP A when there is a function generator connected in series with the device.
Is it possible to get the analyzer to compensate for the presence of the other source. HP ServMan http: Dec 242: The HP A also features high accuracy and error elimination functions to ensure reliable measurements. I want to measure the impedance characteristics of the device when these wire form and compare it to the properties before wire formation. If I actuated from say Hz – 1 kHz using the analyzer, it would change the properties of the device. How reliable is it? Input port and input output port declaration in top module 2.
Synthesized tuning, Part 2: If so, you can use a superimposed DC bias and measure the impedance. Message 13 of I have a better idea now, but I am still not real clear on what you are doing.
Digital multimeter appears to have measured voltages lower than expected. Message 15 of If its circuit really good readable is? Message 19 of